Glove Box Every Day Maintenance Guide
one) Fill in the use report in time just after working the glove box, including "user, drinking water oxygen value, what to do"
two) When changing gas cylinders in the glove box, the strain is established to the default price ± 3mbar
three) The daily doing work pressure is established according to the functioning ease and comfort. When not in use for a extensive time, the strain is set to 1-6mbar
four) Often look at the put on of the gloves, examine the oil degree of the vacuum pump (in between the upstream and downstream markings), and add the pump oil correctly
five) Clean the water probe at the time each 2 to three months to assure precise detection. (Also need to refer to the actual circumstance of the natural environment in different user containers
six) If you frequently use organic solvents in the glove box, it is recommended that you often switch the natural solvent adsorbent materials, not later than one calendar year at the most up-to-date
7) Swap the filter component routinely, no later on than two several years
8) In accordance to the real use of your glove box, make a decision to change the vacuum pump oil cycle, such as black and yellow pump oil, water or more natural solvents in the pump oil. The oil replacement cycle should not exceed 2 yrs at the hottest
9) If a huge total of organic solvent is inevitably exposed all through the experiment, the cycle requirements to be stopped for the experiment. Immediately after the experiment, first thoroughly clean and then start the cycle
10) When the ecosystem in the box is poisonous or hazardous fuel, the exhaust port of the vacuum pump, the exhaust port of the regeneration fuel and the exhaust port of the cleansing valve really should be exhausted to the outside with an exhaust pipe (when exhaust is not allowed to be outside, gloveboxes the exhaust gas cure ought to be done first).